Selective excitation enables encoding and measurement of multiple diffusion parameters in a single experiment
Band selectivity to address specific resonances in a spectrum enables one to encode individual settings for diffusion experiments. In a single experiment, this could include different gradient strengths (enabling coverage of a larger range of diffusion constants), different diffusion delays, or different gradient directions (enabling anisotropic diffusion measurement). In this report, a selective variant of the bipolar pulsed gradient eddy current delay (BPP-LED) experiment, enabling selective encoding of three resonances, was implemented. As proof of principle, the diffusion encoding gradient amplitude was assigned a range dependent on the selected signal, thereby allowing the extraction of the diffusion coefficient for water and a tripeptide (Met-Ala-Ser) with optimal settings in a single experiment.
There is little dispute over the importance of diffusion as a physical effect and its influence on many natural processes.
Diffusion poses a limiting factor in many industrial processes, such as the mixing of chemical reagents to achieve a specific product with sufficient yield. The interplay between intrinsic diffusion rates and operational time-dependent parameters needs to be properly understood and carefully considered in designing such manufacturing processes. Nuclear magnetic resonance (NMR) has proven to be uniquely capable to extract diffusion constants for a specific chemical substance and, in favourable cases, even mixtures, and also to study and understand the processes of diffusion at all length scales and within all compartments available to the molecule under consideration. Thus, NMR has facilitated experimental proof of Fick's laws and has facilitated precise measurements of anisotropic effects arising from spatially extended molecules in solution and subtracting the effects due to geometry, or ionic charges, and the like.
In magnetic resonance imaging, diffusion is currently the limiting spatial resolution factor for inductive NMR detection because excited spins in a molecule tend to diffuse away from their excitation site while waiting for pulse readout. Whereas this uncertainty cannot be removed completely, shorter pulse sequences have the effect of improving resolution by reducing the distance a spin ensemble can wander.
Spatial resolution is especially important in such areas as brain science and brain diagnostics, where anisotropic diffusion tensor imaging is the primary noninvasive means to discover the subcellular structure of a specific brain (Göbel-Guéniot et al., 2020). Essentially, diffusing spin ensembles do not readily traverse cell walls and axons, so that the local structure of brain tissue renders an anisotropic response, with directional weighting being in favour of gliding along the cell walls. In this way, diffusion has helped to reveal brain connectivity patterns. But, also in studies of nanoporous materials, anisotropic diffusion parameters have been used to reveal local nanostructure.
Diffusion anisotropy measurements aim to reveal the 3-dimensional pattern of molecular movement. Mathematically, the movement at a spatial point has to be resolved in three orthogonal directions, which typically would require at least three motionally sensitive measurements, each in combination with a spatial gradient aligned with the specific direction of measurement. The most time-consuming part of the pulse sequence is the echo time and the recovery time. Spatial encoding will, additionally, require the three measurements for each spatial voxel, further slowing down the acquisition.
A degree of freedom available to spectroscopic diffusion measurements is the spectral dispersion, taking advantage of the individual resonances as a means of encoding additional information within a single experiment. This is particularly interesting for molecular mixtures whose components vary in the physical dimensions and, thus, their diffusion coefficient, or for a tracer molecule in an anisotropic environment, where diffusion is dependent on direction. Instead of performing multiple diffusion measurements, each optimized for a particular regime, a frequency bandwidth could be selectively encoded with appropriate diffusion parameters independent of another, different bandwidth, which is itself encoded with differing diffusion parameters. This could be extended down to encoding individual resonances in a spectrum with diffusion parameters appropriate for extracting diffusive properties with high precision.
There are two additional benefits to integrating selective elements to the experiment: first is the elimination of dominating, uninteresting signals (typically solvent) to improve sensitivity to minor components. This has been demonstrated for protein diffusion in water using a selective version of the stimulated echo experiment (Yao et al., 2014) and a selective version of the spin echo experiment for measuring minor components in a mixture (Howe, 2017). The second benefit, noted by Yao et al. (2014) and Howe (2017), is to reduce spectral congestion and, thereby, improve diffusion ordered spectroscopy (DOSY) data analysis, an active field of research in extracting diffusion coefficients from overlapping signals (Aguilar et al., 2010; Colbourne et al., 2011; Foroozandeh et al., 2018; Lin et al., 2020).
In this report, we have extended the stimulated echo experiment using bipolar gradients and longitudinal eddy current compensation (BPP-LED; Wu et al., 1995) to enable selective diffusion encoding. As proof of concept, we demonstrate the ability to selectively address up to three individual resonances (each with a bandwidth of 60 Hz), each encoded with independent diffusion parameters.
In this paper we are specifically paying tribute to Geoffrey Bodenhausen, a pioneer of NMR spectroscopy, on the occasion of his 70th birthday. Our chief source of inspiration comes from one of his earliest papers, published in 1976 (Bodenhausen et al., 1976) while he was at Oxford, which explores selective excitation, and many other of his papers that have explored the measurement of diffusion by various ingenious means.
Deuterium oxide (D2O) and the tripeptide Met-Ala-Ser (MAS) were purchased from Sigma Aldrich and used as received. All H2O was of MQ quality. An NMR sample in a 5 mm NMR tube was prepared by dissolving 10 mg MAS in 500 µL of 90:10 D2O : H2O.
2.2 NMR spectroscopy
NMR experiments were performed using an AVANCE III 500 MHz wide bore NMR spectrometer (Bruker BioSpin GmbH, Ettlingen, Germany) using a Micro5 micro-imaging probe equipped with a 5 mm NMR detector. The sample temperature was controlled using cooling water flowing through the imaging gradient sleeve, and it was maintained at 20 ∘C.
Non-selective diffusion experiments were performed using the bipolar gradient longitudinal eddy current compensated experiment (BPP-LED; Wu et al., 1995, Bruker pulse program ledbpgp2s1d). The diffusion-encoding gradient pulse was the smoothed square (SMSQ10.100), with 2 ms length, 200 µs gradient recovery time, and gradient strength was varied using the parameter optimization (popt) function in TopSpin 3.6.3 (Bruker BioSpin GmbH, Ettlingen, Germany) from 1 % to 95 % of the maximum gradient strength through 16 experiments, and the diffusion time was 250 ms.
Selective diffusion experiments were performed using a modified version of the non-selective experiment (Sel-BPP-LED); 90∘ hard pulses were replaced by sinc-shaped pulses, and 180∘ refocusing pulses were replaced by Gaussian refocusing pulses (Sinc1.1000 and Gaus1_180r.1000 from the Bruker shaped pulse library; Fig. 1). The selective pulse length and power were calculated using the Shape Tool (TopSpin 3.6.3), using a selective bandwidth of 60 Hz, resulting in excitation pulses of 26.8 ms and refocusing pulses of 14.7 ms. The diffusion time was Δ=250 ms, with the timing in the pulse program modified by taking into account the additional selective blocks in the pulse sequence. In total, three signals were chosen for the selective diffusion experiment, namely the water signal (4.8 ppm – parts per million) and, from the MAS peptide, the methyl signal of methionine (2.1 ppm) and the methyl signal of alanine (1.45 ppm). As in the non-selective experiment, the popt function was used to vary the maximum gradient amplitude over 16 experiments. The simultaneous optimization feature of the popt function was used to control the three gradient amplitudes, gn, individually (Fig. 1).
All NMR experiments were performed using radiofrequency (RF) pulses calibrated on the day of measurement. Each measurement consisted of eight scans, with each scan containing 32 K data points over a spectral width of 20 ppm. The free induction decays (FIDs) were zero filled by a factor of 2 and multiplied by an exponential function equivalent to 0.3 Hz line width prior to Fourier transformation. The series of 16 spectra from each diffusion measurement was extracted using an in-house MATLAB script (MATLAB, 2020b; MathWorks, Natick, MA, USA). The MATLAB fit function was used to extract the diffusion coefficients. We note that the DOSY functionality available in TopSpin was neither used in setting up the experiments nor in the data analyses.
Diffusion coefficients were extracted by fitting the intensity versus applied gradient strength using the following equation (Wu et al., 1995; Sinnaeve, 2012):
where I is the normalized NMR signal intensity, γ is the 1H gyromagnetic ratio, δ is the diffusion gradient pulse length, Δ is the diffusion time, τ is the time between the bipolar gradient pulses, and G is the applied gradient strength. It is noted that the value of τ becomes significant in the selective experiment (14.9 ms in this work), and that the equation would need to be specified separately for each selective frequency in case different selective pulse bandwidths, and, therefore, different refocusing pulse lengths, are to be used. There was no correction applied for using the smoothed square gradient shape which, it should be noted, deviates slightly from an ideal rectangular pulse (Sinnaeve, 2012). The diffusion of water in the D2O : H2O mixture was used to calibrate an apparent gradient strength assuming (i) a diffusion coefficient m2/s (taken for pure H2O; Tofts et al., 2000; Holz et al., 2000) and (ii) rectangular gradients. The apparent gradient strength was determined to be 260 G cm−1.
2.3 Results and discussion
A series of experimental NMR spectra measured, using both BPP-LED and Sel-BPP-LED, is presented in Fig. 2, with a summary of the details provided in Table 1. Comparing the signal-to-noise ratio (SNR) when using the BPP-LED vs. Sel-BPP-LED experiments, the water signal decreased by 33 %, while the MAS methyl signals of methionine and alanine were essentially unchanged (−4 % and +2 %, respectively). The Sel-BPP-LED experiment yielded intensity decay curves faithfully reproducing the non-selective experiment (Fig. 3a). A slight variation in the extracted diffusion coefficients was observed (3.3 %, 3.9 %, and −1.5 % relative deviation for Dwater, DMAS,Met and DMAS,Ala). The likely source for the deviations is selective RF pulse imperfections, which could be compensated by using selective pulse shapes with better performance, for example Gaussian pulse cascades Q3 and Q5 (Emsley and Bodenhausen, 1992) or optimal control derived selective pulses (Matson et al., 2009). In the case of all three signals being exposed to 1 %–95 % of gradient maximum, the extracted water diffusion coefficient was m2 s−1, a value with larger fit error (1.1 % vs. 0.37 %) and slightly larger deviation relative to the non-selective experiment (4.0 % vs. 3.3 %) compared to using a gradient strength spanning 1 %–60 %. The extracted values for the MAS Met and Ala methyl signals were essentially unchanged ( m2 s−1 and m2 s−1).
The increased degree of experimental freedom enabled by Sel-BPP-LED is demonstrated in Figs. 2b–c and 3b. Figure 2b reveals that the water signal fully decays well before the MAS when using the same gradient amplitude across all signals. Adjusting the gradient amplitude range for the water signal to apply 1 %–60 % of maximum strength was found to be sufficient to reach full decay at the end of the experiment. In the same experiment, the MAS signals were simultaneously exposed to gradient amplitudes ranging from 1 %–95 % of maximum strength, which was appropriate to sample the intensity decay curve for the larger molecule. As a result, the Sel-BPP-LED experiment enables parallel diffusion measurements of molecules of vastly different diffusive properties by selectively encoding the appropriate diffusion experimental parameter (i.e. gradient amplitude in this case) into a well-resolved signal.
Selective variants of numerous NMR experiments have been realized since the idea of converting multi-dimensional to 1D experiments using selective pulses was reported (Kessler et al., 1986). Such experiments are often exploited to simplify otherwise complex spectra (Kiraly et al., 2021; Alexandersson et al., 2020; Poggetto et al., 2017; MacKinnon et al., 2016; Pelupessy et al., 1999), to suppress strong signals that dominate the dynamic range (Cutting et al., 2000), or to selectively drive desired coherence pathways (Haller et al., 2019; Ferrage et al., 2004; Chiarparin et al., 1998). Diffusion experiments have also benefited from the addition of selectivity to reduce the influence of strong solvent signals in biomolecular samples (Shukla and Dorai, 2011; Yao et al., 2014), to focus the diffusion experiment onto select molecules in a complex mixture (Lyu et al., 2018), or to use the diffusion dimension as an access point to further selective-based NMR experiments (e.g. selective TOCSY – Total Correlation Spectroscopy; Lyu et al., 2018, and supplemented with relaxation encoding; Poggetto et al., 2017).
The selective experiment described in this report offers the same advantages in terms of signal suppression and reducing complexity. A key difference compared to previous selective diffusion reports is the encoding of information into individual resonances. This is in contrast to the method described by Howe (2017), who demonstrated diffusion measurements by the selective excitation of four resonances. Since every resonance was exposed to the same experiment parameters, all four resonances could be simultaneously excited resulting in a net gain in time per scan compared to the experiment described here. Yao et al. (2014) demonstrated a selective diffusion experiment using selective pulses of larger bandwidth (∼ 2 kHz) and, therefore, also gained in experimental time per scan. The benefit of the Sel-BPP-LED experiment (in fact, we are not limited to BPP-LED; any stimulated echo based-diffusion experiment should be compatible with the concept) is the ability to parallelize measurements, thus accessing diffusion information from multiple sample species in a single experiment.
The ability to accelerate measurements via parallelization will be important in many applications. For example, considerable advantages for increased measurement time efficiency for DTI (diffusion tensor imaging) applications are anticipated, since the 3D diffusion tensor (with up to six components) needs to be determined on a voxel-by-voxel basis. In this case, the ability to encode up to six scalar components in a single k-space acquisition, i.e. in one shot and subsequent FID, would be a tremendous advantage. For technical systems, this could be achieved by using fluids of designed composition, such that at least six well-resolved signals are available. For clinical DTI, the measurement will be reliant on identifying molecules that are sufficiently abundant and a chemical shift resolution at typical MRI scanner field strengths (e.g. at 3 T, candidates could include lactate, N-acetyl amino acid derivatives, choline derivatives, or creatine derivatives; Wilson et al., 2019); however, even if all six tensor components cannot be encoded, there will still be an measurement acceleration when more than one component can be encoded. An additional example is complex samples containing both slow- and fast-diffusing species where restricted diffusion is present. Using a selective diffusion experiment, the optimal parameters can be simultaneously used for the fast-diffusing species and, in the example presented in Fig. 4, the restricted slow-diffusing species.
The limit to the number of selective units, n, in the selective diffusion experiment is governed by (i) the desired bandwidth to be selectively encoded, (ii) the desired diffusion time Δ, and, related, (iii) the T1 relaxation of the selected signals. For the worst-case scenario, selectivity with small bandwidth of multiple resonances with short T1 will not be possible since the resulting diffusion time would be too long, and signal intensity would be lost to relaxation. In the experiment demonstrated here, a selective bandwidth of 60 Hz was used exclusively, requiring selective pulse lengths of 26.8 ms and 14.7 ms for excitation and refocusing. This limited the diffusion times ms for the three signals selectively encoded or, more generally, . Shorter diffusion times can be accessed by selectively encoding spectral regions instead of individual resonances, thereby reducing the required length of the selective pulse (in the case of J-coupled signals within the excited bandwidth, there will be minor signal loss to anti-phase coherence, which will be destroyed by the gradient applied after magnetization storage). This has been demonstrated for protein and peptide diffusion measurements by selectively exciting bandwidths of the order of 2 kHz with the benefit of relatively short selective pulse lengths of the order of 2.5 ms (Yao et al., 2014). Encoding more than one resonance with the same diffusion parameters would also be possible using pulses with multiband selectivity, which would effectively reduce n and also enable access to shorter diffusion times.
Taking advantage of individual resonances as a means to encode additional, unique information into a single experiment is a means to achieve measurement parallelization. For each unique encoding n, experimental time is accelerated by a factor n. In this work, a Sel-BPP-LED diffusion experiment demonstrated this concept, with n=2 in this case, since two signals belonged to the same molecule and were encoded with the same gradient direction. While demonstrated for the BPP-LED experiment, the concept is general and can be applied to any diffusion pulse sequence implementing the stimulated echo. Selective encoding enables the simultaneous measurement of diffusion coefficients of sample mixture components experiencing different transport properties dictated by molecular size or selective restriction via matrix interactions. This principle could be extended to anisotropic diffusion by encoding gradient directions into different resonances. To improve the experiment, selective RF pulses with better performance should be explored to avoid systematic deviations in the extracted diffusion coefficients.
The Sel-BPP-LED pulse program, tested using TopSpin 3.6.3, and the NMR data are available at https://doi.org/10.5281/zenodo.5105713 (MacKinnon et al., 2021).
PS, MJ, and JGK conceptualized the paper, while NM led the formal analysis and visualization. The investigation was done by NM, MA, and TG. BL, MJ, and JGK supervised the project. JGK and NM wrote the original draft, with all co-authors contributing to the review and editing of the paper.
Some authors are members of the editorial board of Magnetic Resonance. The peer review process was guided by an independent editor, and the authors have also no other competing interests to declare.
Publisher's note: Copernicus Publications remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
This article is part of the special issue “Geoffrey Bodenhausen Festschrift”. It is not associated with a conference.
Jan G. Korvink acknowledges the support from the framework of the German Excellence Intitiative (grant no. EXC 2082), “3D Matter Made to Order”, and, together with Burkhard Luy, the KIT-VirtMat initiative “Virtual Materials Design II”. Jan G. Korvink, Mazin Jouda, Thomas Gloge, Burkhard Luy, and Neil MacKinnon acknowledge the partial financial support of the Helmholtz Association through the programmes “Science and Technology of Nanosystems – STN”, “BioInterfaces in Technology and Medicine – BIFTM”, and “Materials Systems Engineering – MSE”. The authors acknowledge the support of the Karlsruhe Nano Micro Facility (KNMFi), a Helmholtz Research Infrastructure at Karlsruhe Institute of Technology.
This research has been supported by the Horizon 2020 (TISuMR; grant no. 737043), the Deutsche Forschungsgemeinschaft (grant no. KO 1883/39-1), and the European Research Council (grant no. 951459).
The article processing charges for this open-access publication were covered by the Karlsruhe Institute of Technology (KIT).
This paper was edited by Fabien Ferrage and reviewed by Davy Sinnaeve and one anonymous referee.
Aguilar, J., Faulkner, S., Nilsson, M., and Morris, G.: Pure Shift 1H NMR: A Resolution of the Resolution Problem?, Angew. Chem. Int. Edit., 49, 3901–3903, https://doi.org/10.1002/anie.201001107, 2010. a
Alexandersson, E., Sandström, C., Lundqvist, L. C. E., and Nestor, G.: Band-selective NMR experiments for suppression of unwanted signals in complex mixtures, RSC Advances, 10, 32511–32515, https://doi.org/10.1039/d0ra06828d, 2020. a
Bodenhausen, G., Freeman, R., and Morris, G. A.: A simple pulse sequence for selective excitation in Fourier transform NMR, J. Magn. Reson., 23, 171–175, 1976. a
Chiarparin, E., Pelupessy, I., and Bodenhausen, G.: Selective cross-polarization in solution state NMR, Mol. Phys., 95, 759–767, https://doi.org/10.1080/002689798166396, 1998. a
Colbourne, A. A., Morris, G. A., and Nilsson, M.: Local Covariance Order Diffusion-Ordered Spectroscopy: A Powerful Tool for Mixture Analysis, J. Am. Chem. Soc., 133, 7640–7643, https://doi.org/10.1021/ja2004895, 2011. a
Cutting, B., Chen, J.-H., Moskau, D., and Bodenhausen, G.: Radiation damping compensation of selective pulses in water–protein exchange spectroscopy, J. Biomol. NMR, 17, 323–330, https://doi.org/10.1023/a:1008303322609, 2000. a
Emsley, L. and Bodenhausen, G.: Optimization of shaped selective pulses for NMR using a quaternion description of their overall propagators, J. Magn. Reson., 97, 135–148, https://doi.org/10.1016/0022-2364(92)90242-y, 1992. a
Ferrage, F., Eykyn, T. R., and Bodenhausen, G.: Frequency‐Switched Single‐Transition Cross‐Polarization: A Tool for Selective Experiments in Biomolecular NMR, ChemPhysChem, 5, 76–84, https://doi.org/10.1002/cphc.200300905, 2004. a
Foroozandeh, M., Morris, G. A., and Nilsson, M.: PSYCHE Pure Shift NMR Spectroscopy, Chemistry – A European Journal, 24, 13988–14000, https://doi.org/10.1002/chem.201800524, 2018. a
Göbel-Guéniot, K., Gerlach, J., Kamberger, R., Leupold, J., Elverfeldt, D. v., Hennig, J., Korvink, J. G., Haas, C. A., and LeVan, P.: Histological Correlates of Diffusion-Weighted Magnetic Resonance Microscopy in a Mouse Model of Mesial Temporal Lobe Epilepsy, Front. Neurosci., 14, 543, https://doi.org/10.3389/fnins.2020.00543, 2020. a
Haller, J. D., Bodor, A., and Luy, B.: Real-time pure shift measurements for uniformly isotope-labeled molecules using X-selective BIRD homonuclear decoupling, J. Magn. Reson., 302, 64–71, https://doi.org/10.1016/j.jmr.2019.03.011, 2019. a
Holz, M., Heil, S. R., and Sacco, A.: Temperature-dependent self-diffusion coefficients of water and six selected molecular liquids for calibration in accurate 1H NMR PFG measurements, Phys. Chem. Chem. Phys., 2, 4740–4742, https://doi.org/10.1039/b005319h, 2000. a
Howe, P. W.: Selective diffusion spectroscopy using excitation sculpting, Magn. Reson. Chem., 55, 433, https://doi.org/10.1002/mrc.4402, 2017. a, b, c
Kessler, H., Oschkinat, H., Griesinger, C., and Bermel, W.: Transformation of homonuclear two-dimensional NMR techniques into one-dimensional techniques using Gaussian pulses, J. Magn. Reson., 70, 106–133, https://doi.org/10.1016/0022-2364(86)90366-5, 1986. a
Kiraly, P., Kern, N., Plesniak, M. P., Nilsson, M., Procter, D. J., Morris, G. A., and Adams, R. W.: Single‐Scan Selective Excitation of Individual NMR Signals in Overlapping Multiplets, Angew. Chem. Int. Edit., 60, 666–669, https://doi.org/10.1002/anie.202011642, 2021. a
Lin, E., Yang, Y., Huang, Y., and Chen, Z.: High-Resolution Reconstruction for Diffusion-Ordered NMR Spectroscopy, Anal. Chem., 92, 634–639, https://doi.org/10.1021/acs.analchem.9b03865, 2020. a
Lyu, Z., Yue, F., Yan, X., Shan, J., Xiang, D., Pedersen, C. M., Li, C., Wang, Y., and Qiao, Y.: Combination of DOSY and 1D selective gradient TOCSY: Versatile NMR tools for identify the mixtures from glycerol hydrogenolysis reaction, Fuel Process. Technol., 171, 117–123, https://doi.org/10.1016/j.fuproc.2017.11.011, 2018. a, b
MacKinnon, N., While, P. T., and Korvink, J. G.: Novel selective TOCSY method enables NMR spectral elucidation of metabolomic mixtures, J. Magn. Reson., 272, 147–157, https://doi.org/10.1016/j.jmr.2016.09.011, 2016. a
MacKinnon, N., Alinaghian, M., Silva, P., Gloge, T., Luy, B., Jouda, M., and Korvink, J. G.: Selective excitation enables encoding and measurement of multiple diffusion parameters in a single experiment (1.0.0), Zenodo [data set], https://doi.org/10.5281/zenodo.5105713, 2021. a
Matson, G. B., Young, K., and Kaiser, L. G.: RF pulses for in vivo spectroscopy at high field designed under conditions of limited power using optimal control, J. Magn. Reson., 199, 30–40, https://doi.org/10.1016/j.jmr.2009.03.010, 2009. a
Pelupessy, P., Chiarparin, E., and Bodenhausen, G.: Excitation of Selected Proton Signals in NMR of Isotopically Labeled Macromolecules, J. Magn. Reson., 138, 178–181, https://doi.org/10.1006/jmre.1999.1715, 1999. a
Poggetto, G. D., Castañar, L., Adams, R. W., Morris, G. A., and Nilsson, M.: Relaxation-encoded NMR experiments for mixture analysis: REST and beer, Chem. Commun., 53, 7461–7464, https://doi.org/10.1039/c7cc03150e, 2017. a
Shukla, M. and Dorai, K.: Resolving overlaps in diffusion encoded spectra using band-selective pulses in a 3D BEST-DOSY experiment, J. Magn. Reson., 213, 69–75, https://doi.org/10.1016/j.jmr.2011.08.038, 2011. a
Sinnaeve, D.: The Stejskal-Tanner equation generalized for any gradient shape-an overview of most pulse sequences measuring free diffusion, Concepts Magn. Reson. A, 40A, 39–65, https://doi.org/10.1002/cmr.a.21223, 2012. a, b
Tofts, P., Lloyd, D., Clark, C., Barker, G., Parker, G., McConville, P., Baldock, C., and Pope, J.: Test liquids for quantitative MRI measurements of self‐diffusion coefficient in vivo, Magn. Reson. Med., 43, 368–374, https://doi.org/10.1002/(sici)1522-2594(200003)43:3<368::aid-mrm8>3.0.co;2-b, 2000. a
Wilson, M., Andronesi, O., Barker, P. B., Bartha, R., Bizzi, A., Bolan, P. J., Brindle, K. M., Choi, I., Cudalbu, C., Dydak, U., Emir, U. E., Gonzalez, R. G., Gruber, S., Gruetter, R., Gupta, R. K., Heerschap, A., Henning, A., Hetherington, H. P., Huppi, P. S., Hurd, R. E., Kantarci, K., Kauppinen, R. A., Klomp, D. W. J., Kreis, R., Kruiskamp, M. J., Leach, M. O., Lin, A. P., Luijten, P. R., Marjańska, M., Maudsley, A. A., Meyerhoff, D. J., Mountford, C. E., Mullins, P. G., Murdoch, J. B., Nelson, S. J., Noeske, R., Öz, G., Pan, J. W., Peet, A. C., Poptani, H., Posse, S., Ratai, E., Salibi, N., Scheenen, T. W. J., Smith, I. C. P., Soher, B. J., Tkáč, I., Vigneron, D. B., and Howe, F. A.: Methodological consensus on clinical proton MRS of the brain: Review and recommendations, Magn. Reson. Med., 82, 527–550, https://doi.org/10.1002/mrm.27742, 2019. a
Wu, D., Chen, A., and Johnson, C.: An Improved Diffusion-Ordered Spectroscopy Experiment Incorporating Bipolar-Gradient Pulses, J. Magn. Reson., 115, 260–264, https://doi.org/10.1006/jmra.1995.1176, 1995. a, b, c
Yao, S., Weber, D. K., Separovic, F., and Keizer, D. W.: Measuring translational diffusion coefficients of peptides and proteins by PFG-NMR using band-selective RF pulses, Eur. Biophys. J., 43, 331, https://doi.org/10.1007/s00249-014-0965-x, 2014. a, b, c, d, e